OVERVIEW AND TECHNICAL FEATURES
The Pressure Reducing Manifold can be installed inside an hyperbaric chamber and can be used for air and Sat, to supply gas to the BIBS.
The primary function of this pressure reducer manifold as it has been designed is to allow operators in a hyperbaric chamber to have a reducer dedicated to every single BIBS. Otherwise, normally hyperbaric chambers are equipped with a reducer on the main control panel and a reducer for each hyperbaric chamber lock which will provide the supply to all BIBS contains in the lock itself.
Through the use of this pressure reducing manifold it can be removed the single driven reducer positioned externally to the lock avoiding the single point of failure. During the study of this new pressure reducing manifold there have been taken into consideration other possible system faults, see the rupture of one of the BIBS. In the traditional systems all the BIBS are installed on one manifold with all supply connections between them. We have studied a manifold system which allows the removal and/or the exclusion of one of the masks without the necessity of completely closing the supply to all, in order to take action on that one eventually damaged and change it.
This reducer is made in Brass and the finish surface is chrome polished. The reducer is made of oxygen compatible materials and so are the seals, and the grease used for lubrication is approved by BAM for oxygen use.
The HM-01 can be configured as per client need, but the standard are:
Compliant with Standard PED 97/23/EC.
Note: The pressure reducer are NOT included in the price. Selecting the number of port you will configure only the manifold with the valve.
Please see attached interface drawing